Practical HPLC Troubleshooting Video
Real Life HPLC Cycling Baseline problem solved live on video using the
CHROMacademy HPLC Troubleshooter
In this video, we solve a cycling HPLC baseline problem submitted to our Ask the Expert service by a CHROMacademy member. This video shows Tony Taylor (Technical Director, Crawford Scientific) explaining how to use the interactive HPLC Troubleshooter to identify potential causes for the problem. He then discusses each of the potential causes, how they might affect the appearance of our baseline and what to do to avoid the problems happening again! Tune in to see what eventually fixed the problem!